Background and aim: Current methods used to take impressions for custom foot orthoses include plaster bandage, foam box, fiberglass, and laser-optical scanner. Impressions are converted to positive plaster or foam models. These methods create waste and may not be feasible in resource-limited areas. This technical note presents an alternative, greener impression and fabrication technique for foot orthoses that utilizes the dilatancy principle. Technique: Steps of the dilatancy (vacuum-based) procedure include taking an impression of the foot, converting the negative mold to a positive sand model, modifying the positive sand model, and thermoforming the foot orthosis. Discussion: This plaster-less system is inexpensive to set up and maintain, is reusable thereby minimizing cost and waste, and is clean to use. It enables a practitioner to quickly take an impression for fabricating a foot orthosis in a short period of time during a single clinic visit by the patient.
ASJC Scopus subject areas
- Health Professions (miscellaneous)